超级彩虹荧光微球，UltraRainbow 荧光微球,3.0-3.4 um
Brand : Spherotech
Instructions for use:
A. Preparation of Particles
1. Vortex the particles vigorously
2. Add 2 to 4 drops of particles to 1mL of sheath fluid or DI water. The inclusion of a small amount of detergent (~0.01%) in the dilution buffer will increase the percentage of the singlet population.
B. Daily Alignment
To determine the optical alignment of the system perform the following:
1. Set a live gate for the singlet population on the FSC vs SSC histogram to exclude aggregates
2. Adjust the Gain and High voltage so that the mean channel number of the peak is in a predetermined position on each histogram of interest. The histograms on the previous page
can be used as a guide.
3. Collect 5000 events inside the gate
4. Record the % CV and High Voltage for all fluorescence channels of interest.
5. Use a computer program such as Excel to generate the Levy Jennings graphs.
NOTE: If the values on any parameter exceed those of the day-to-day average or preset values, which are determined by at least one months worth of data, additional calibration or alignment procedures should be performed according to the instrument operation manual.
URFP-30-2, 单峰荧光微球, 校直微球, 3um荧光球, 流式细胞仪校准